Crisis like Flint and Newark highlight the lack of effective large-scale water quality assurance and the huge human cost this has. Current testing options such as lab testing and test strips are expensive, single use and heavily rely on expertise. Since water quality changes throughout the day and the seasons, multiple tests are necessary for a real assessment of heavy-metal contamination. Thus, there is to date no test that can help answering the question: is there lead in my water?
The (idea for a) Solution
**Note this are just my initial research notes, far from a finalized working device. Wanted to share anyway even after a summer with limited resources (thanks Covid) to gauge interest, get feedback and possibly find collaborators**
**Highly recommend familiarizing yourself with the WheeSat and the Standard Additons Method before reading further**
Deployable, inexpensive, quick and accurate are the characteristics that makes analytical electrochemistry the branch of science that can perhaps provide an answer. Digital, inexpensive potentiostats like the WheeSat (used as base for our project, shootout to Prof. Summers for being so awesome!) have proven to be effective tools at an extremely low price-point. Additionally, (almost) ready-to-go screen printed electrodes ('SPEs') (such as https://www.palmsens.com/product/italsens-is-hm/) virtually eliminate the need for electrode preparation and conditioning thereby eliminating yet another technical hurdle. However, there is one last challenge remaining: Anodic Strip Voltammetry ('ASV', the technique used to measure chemicals like lead) isn't really a term that comes up in every day parlance so in order for it to be effective, it has to be more "user friendly": that is, doable even if you don't have a degree in electrochemistry. That is where our work and cutting-edge advances in embedded-AI come in. The idea is the following:
By turning the problem of parameter optimization for generation of voltammograms (the experimental part" of ASV) into an ML problem, we can leverage pre-existing hardware and software solutions in a way that the need for electrochemical knowledge is virtually eliminated.
Turning electrochemistry problem into an ML problem
We base our work partying from the standard-addition method followed by a two-step strip voltammetry protocol. The two steps are:
Step 1. Pre-concentration (a very simple set-up than can be easily automated), biasing the working electrode to +5V as and pre-concentrating for several minutes. This potential is more than enough for all possible species detectable with SV and SPE, including lead, mercury, arsenic (see https://link.springer.com/article/10.1007/s00604-015-1651-0/tables/3)
Step2. Strip Voltammetry.
The idea here is that the voltammogram (the plot of interest) generated will start from pre-set parameters, and then the resulting (potentially "bad") voltammogram will be categorized based on features (things like shape, angle, patterns in values, etc) determined by the pre-trained ML model. After being categorized, new, better experimental parameters will be decided, and the experiment will be run again until a "good" voltammogram is achieved.
Machine Learning for this Application
While writing a custom ML model for this application could certainly be an option, the "smart" way is to implement and adapt open-source pre-existing ML models which for which we then simply provide the training data set. By pre-training we can also minimize the hardware requirements to the bare minimum which improves deployability and cost. We consider various Google and TensorFlow available APIs, for which the latter has some promising computationally inexpensive embedded solutions.
A big reason why we believe this can work even with little training is because of the sampling space. While water is an extremely complicated matrix with parameters that vary widely, the behavior changes "predictably" and we know enough about the main parameters to consider: conductivity, pH and temperature. Consider the following Pourbaix Diagram:
By far, the largest question remainin is: how many data points do we need to train the model? While some experiments are still necessary to provide an answer, the hope is that "not too many" are needed, and by implementing different, improved ML models the goal is to keep that number low, with perhaps a few hundred water samples needed to sample the whole space. The effect of instrumental drift also remains to be determined and dealt with.
A vague statement made above is that the sequential experiment will stop once a "good" voltammogram is achieved. This is a classic basic problem in computer vision in which a match index can be easily computed. In general, you can ask imaging software (like imageJ) to tell you the percentage match between two shapes. Since a "good" voltammogram presents a bell curve at each species of interest (at given potentials), we can implement this solution in an straightforward manner.
Improved Deployment: Inexpensive and wireless
One of the main reasons why we decide to take this embedded-AI approach was because under this scheme we don't need the deployed system to carry heavy computations and can therefore work with a very basic, inexpensive "brain". We pre-train the model and then we load it onto the device, and it is then ready to go!
Now that we were working with the RaspberryPi system we can take the next step and leave behind our computers and just take our phones. As the python code is loaded in the RaspberryPi, we created an Android app which connects to the Raspberry Pi and the has a button to execute the main.py and then the results are spat our in the screen of your phone. Neat! Here are the files and this is the post with the instructions.
The repository also includes a mock app ('barebones', apk file) that 'talks' with the Raspberry, in the future that is the type of interface we would like to use. This mock app only works for Android (for now) and it was developed on a Linux environment (app development isn't that friendly for your information) so may not be the easiest to adapt/implement.
What a fascinating application of machine learning (using existing models, to boot!), ultimately making this tool more accessible. Thanks for explaining it all!
Thank you for your comments, hope we get to collect enough data to train a model enough to make it useful in real samples.
Reply to this comment...
Log in to comment
I found your project proposal very interesting.
I have no practical experience with Electrochemistry but I am studying the subject with plans to assemble a potentiostat and use it in projects for chemical analysis of water.
Translated by Google:
I would like to learn more to "try to contribute" to your project.
Can I use this list for exchanging messages? Or do you prefer use the project page?
Best Regards, Markos
Is this a question? Click here to post it to the Questions page.
Markos, thanks for your interest! If you're wanting to explore potentiostats I'd recommend looking at the WheeSat (by Prof. Jack Summers and Smoky Mt. Scientific) which is what we're using. If you have interest in embedded AI applications then we could talk and see if there's a way for this to fit with your studies.
Reply to this comment...
Log in to comment
Login to comment.