This is a draft -- still in progress, based on this note
I'm still working on:
- a graph of replications (could be added later)
- references, from https://publiclab.org/wiki/spectrometer-curriculum
One thing spectrometers can be used for is to detect gases, but an experimental setup for this is actually pretty hard, because gases are so transparent. Think about how little light is blocked by the air when you can look across a mountain valley.
How much of a given gas would you need to look "through" to actually see certain colors blocked -- say, if you looked at a full spectrum (incandescent or halogen) light bulb with your spectrometer? Turns out, the width of the Earth's atmosphere is more than enough.
In this activity, we'll just point a spectrometer at the sky to (ideally) see iron, calcium, hydrogen, sodium, as well as water and C02, as dark "blocked" bands in the spectrum of the sun, which would otherwise be smooth and continuous. (above image by @liz, at #ecohack)
Activities to do first
What you'll need
- a Desktop Spectrometry Starter Kit or a foldable mini-spectrometer
- a laptop and a window, or an outdoor space to work
- an account on SpectralWorkbench.org
- two people are really helpful -- one to point the spectrometer, one to operate the laptop
You can read more about the Fraunhofer lines, why they're called that, and why the sun makes a good full-spectrum light source in @cfastie's great post here, entitled Fraunhofer.
Photo by @donblair - pointing a DIY spectrometer from the roof of a building.
If you haven't recently, calibrate your spectrometer again to ensure it hasn't drifted.
Find a spot outside or on a window sill where you can point your spectrometer at the sky -- not at the sun directly (that's too bright) but at a cloud, or at the blue part of the sky.
It's helpful if you can affix the spectrometer, like with a tripod, or rubber banding or zip-tying it to something pointable, like a microphone stand or even a desk lamp. Having a second person to "manage" the pointing is really helpful while you're using the laptop.
Go to https://spectralworkbench.org/capture, plug in your spectrometer, and capture a spectrum.
Copy your last calibration with the
Copy calibration button. Tag your spectrum with:
fraunhofer so we can find it!
Bonus: tweet out a picture of yourself pointing your spectrometer at the sky!
Post links to your spectra here so we can see how consistently people are able to find these. Please note what the weather was like, what time of day it was, and anything else that might
Adding a photo of yourself (like the lead image of this post) to show your setup and the weather is a great way to show your experimental setup. It'll help with troubleshooting, and help others see how to do this activity themselves.
The sun is so bright!!! It's easy to overpower your camera and cause clipping -- where brightness in any of the three color channels of the camera reaches 100% and the top of the graph just gets flattened out. Try pointing further away from the sun, or even the sun's reflection off a piece of white paper (though this may change your spectrum!).
Direct vs. diffuse sunlight may yield different spectra, as explored in this note. An example of clipping is here, in the blue area to the left, where the brightness curve gets flattened against the top:
What outcome you should get
Post your results in a response below -- you should be able to see some faint black bands through your spectrum as in this image:
See many other peoples' attempts at this here -- some are overexposed in the visible range, but have interesting absorption lines in the infrared range: https://spectralworkbench.org/tags/sunlight
Here's one which shows clear black lines in the infrared range, to the right side; if you remove the infrared blocking filter from your webcam, you can record these wavelengths too:
Login to comment.