I am a new user trying to make use of the mini-foldable spectrometer with my phone. The images I ...
Public Lab is an open community which collaboratively develops accessible, open source, Do-It-Yourself technologies for investigating local environmental health and justice issues.
Show your support for community science. Donate to Public Lab »
All topics »
If you cannot use the ReCaptcha to verify you are not a bot, use this alternative verification.
As an open source community, we believe in open licensing of content so that other members of the community can leverage your work legally -- with attribution, of course. By joining the Public Lab site, you agree to release the content you post here under a Creative Commons Attribution Sharealike license, and the hardware designs you post under the CERN Open Hardware License 1.1 (full text). This has the added benefit that others must share their improvements in turn with you.
sign up to join the Public Lab community
Forgot your password? Reset it here
by chuynh451 |
November 01, 2016 20:21 |
I am a new user trying to make use of the mini-foldable spectrometer with my phone. The images I get have very bright lines and yet the intensity is labeled as 7% or lower. How do I get the intensity up?
Can you link to your spectra so I can see?
Is this a question? Click here to post it to the Questions page.
Reply to this comment...
Log in to comment
Here is my latest try: https://spectralworkbench.org/spectrums/89850
Thanks for sharing the link. Because you're using an uploaded image, the top row of pixels is quite dark, and that's the one used as your cross-section line.
I tried forking your spectrum and selecting a new cross-section: https://spectralworkbench.org/spectrums/89900
That looks much better and makes a good calibration. But note that since your spectrum is curved, the choice of cross-section will shift your calibration left or right depending on what height it is. You'll need to ensure all your data has the same cross-section if you want to compare it or use this calibration for other readings.
Hope that helps!
You'll need to ensure all your data has the same cross-section if you want to compare it or use this calibration for other readings.
copying calibration issue
updated about 2 months ago
Spectral Workbench so slow on raspberry pi zero
updated 3 months ago
Reading the individual pixel data from a piCamera sensor sans-lens?
updated 5 months ago
How do I select the back camera of the smartphone?
updated 7 months ago
I just gotten my newly purchased Infragram Pi Camera, my pi.local is blank and pi.local/Cam keeps popping up message "Error in RaspiMJPEG" hence I do not get any live video feed. Anyone can help?
updated 8 months ago
Sorry for asking such a elementary question, what does the intensity % means? Can it shows how strong the light source is? Is it in terms of lux?
Can you calibrate this?
updated 9 months ago
Can you give me mesure for spectrum? Thank you.
updated 10 months ago
This is part of: