Papercraft Spectrometry Starter Kit
New to spectrometry? On a budget? Looking for something to introduce a group to spectrometry? You might like to try the Papercraft Spectrometry Starter Kit, a simple foldable paper spectrometer which you can attach to a smartphone or webcam. It’s made of paper to reduce cost and complexity, and is mainly intended as a low cost “introductory” or educational kit.
The resolution and stability (ability to hold a wavelength calibration) of the spectrometer depend on how carefully you construct and store it. This kit is designed to be easy to assemble, and is not focused on precision or consistent measurements; it’s made of paper, and will crush if you put it in your pocket -- but it can be made out of easy-to-find materials and assembled in ~15 minutes. To improve it, see the related hardware mods and feel free to post your own adjustments as a research note. For a more rigid, robust device intended for more advanced work, check out the Desktop Spectrometry Starter Kit.
Most digital cameras can record light in the visible range, ~400-700 nanometers, so this determines the range of your device. The resolution is mainly limited by the resolution of the camera (and its focus!) and by the narrowness of the slit -- but measurements should be possible at better than 3 nanometers per pixel. See limitations and goals for more.
Have you added to your starter kit, improved it, or redesigned it? Show others how to take it to the next level by posting a build guide here:
|Tweaks to foldable mini-spectrometer design||build||proposal||@warren||1h||easy||1 builds: Try it »|
|3D printed mini spectrometer||-||-||@rthalman||-||-||1 builds: Try it »|
|Reduce light inside foldable PL spectrometer||-||-||@xose||-||-||0 builds: Try it »|
|Silhouette Cameo desktop paper cutter for prototyping||-||-||@warren||-||-||0 builds: Try it »|
|Silhouette cut Public Lab Mobile spectrometer v2||-||-||@briandegger||-||-||0 builds: Try it »|
|Printable 3D Spectometer||-||-||@mdiego2798||-||-||1 builds: Try it »|
Upgrades should include a parts list and a step-by-step construction guide with photo documentation. See an example.
Get involved in the next revision
As these upgrades mature, the Kits initiative will incorporate some into the Foldable Spectrometer Starter Kit itself (depending on cost & complexity) as part of our commitment to an open hardware design process
Frequently Asked Questions
This section is for questions about the Foldable Spectrometer Starter Kit, specifically. For questions about spectrometry in general, see this FAQ.
|I'm completely confused. I built the papercraft one, but then what do I do?||@mactac11||over 2 years ago||0||1|
|Split DVD – which part to use for spectrometer||@Nelari||over 3 years ago||0||4|
|Can you give me measure for spectrum? Thank you.||@Skorex||over 3 years ago||0||2|
|Can I get a file of numbers instead of a graph from the foldable spectrometer software?||@mountevans||over 4 years ago||1||4|
|Do I need to use a DVD-R? Will a CD-R work too?||@jordankc||over 4 years ago||1||2|
|Paper for printing||@G33K4P00RV4||almost 5 years ago||0||5|
|Noise when Trying to Analyze Spectrum||@matthart610||almost 5 years ago||0||1|
|Can I use a Diffraction Grating Slide Holographic 1000 lines/mm instead of the CD?||@oglezsa||almost 5 years ago||1||3|
|What's an easy way to compare two liquid samples with a spectrometer?||@warren||almost 5 years ago||1||7|
Plans, materials, and what it takes to make one:
- print and/or trace this letter-sized PDF pattern onto black paper
- cut out with scissors, no other special equipment required
- fold the paper along the lines to form a rectangular tube (takes about 15 minutes)
This starter kit includes these materials:
- a piece of pre-printed paper (dimensions) with a hole for the camera
- a DVD-R
- a narrow slit printed onto plastic film (see image - and how to make one of these)
The kit uses, but does not include, a smartphone with a camera (or a webcam) to capture images of spectra through the spectrometer you construct. It works with free web software https://spectralworkbench.org.
See how to get started using this kit in the Activities section of the Spectrometry page.
Limitations and Goals
This starter kit is a great way to begin learning about spectroscopy by building a simple one, but is not meant for quantitative research. Once you’ve mastered it, you can improve it with some of the hardware mods listed here, or to take it to the next level, see the Desktop Spectrometry Starter Kit for a more robust platform to build on.
Though it is specifically designed to be an “introductory” kit, there are a number of goals to work towards, and limitations which you can help to overcome in the current design. Intro-level kits play an important role in helping people build the skills they need to join in the project and improve these methods -- in building a large collaborative community around this project.
Some of the goals and challenges we’re still working on with this kit include:
- the alignment between the foldable spectrometer and a smartphone or webcam is hard to keep consistent with soft materials like paper
- assembly relies on careful gluing or taping -- could we make a glue-less tab-based version that’s still light tight?
- the film-printed slit in the starter kit is really nice, but can be hard for someone to make on their own -- some tips on making your own slits would be helpful
- a wider spectrometer would make a wider spectrum, which would be easier to capture a picture of
- comparing spectra with different light sources, or between different devices, is not well explored; learn more this challenge here
Get involved in the next revision
To make a diffraction grating from a DVD-R, peel back the reflective layer, and cut a small square out of the transparent layer. See also: Preparing a DVD-R to act as a diffraction grating for more information on the optional removal of the residual aluminium and dye layers.
Illustrated Assembly Guide
Cut out along the outer edge if not already cut out.